Our devices MC-780, DC-430, DC-360 and BC-601 have the option to store measurement data on an SD-card. The data on this SD-card can be read by inserting the SD-card in a PC or laptop. When TANITA software is installed on this device the data can be imported in the software to generate reports.
When the SD-card gets damaged the BC-601 can show an error message ("err") when it is inserted in the device. This problem can be solved by replacing the SD-card.
It is important that the SD-card has a maximum capacity of 16 GB. The device will not work with SD-cards that have more storage capacity than 16 GB. The logo's on these type of SD-cards (SD/ SDHC) look like:
SD-cards with the logo's below usually have more storage capacity and do NOT work with the device (SDXC/ SDUC):